Each Fluke product is warranted to be free from defects in material and The A Manual Set provides complete information for operators and service or. Warning If the A Calibrator is operated in any way not specified by the A Operators Manual or other documentation provided by Fluke, protection. View and Download FLUKE A operator’s manual online. Multi-Product Calibrator. A Test Equipment pdf manual download.
|Published (Last):||6 February 2014|
|PDF File Size:||1.80 Mb|
|ePub File Size:||2.24 Mb|
|Price:||Free* [*Free Regsitration Required]|
AC Voltage amplitude Procedure provided manul this manual. Thermocouple Measurement Accuracy The Thermocouple Measurement Accuracy test checks the internal temperature reference.
Verify that the calibrator is in standby and connect the DMM as shown in Figure The Control Display now shows your entries.
The front panel calibration is self-prompting and takes you through the complete procedure. Both negative and positive values must be entered.
You have several choices for sending commands to the A: Then follow these steps to for each period listed in Table Page 62 A Operator Manual You can also redirect A current to the A output terminals; 0 – 2. AC Current Calibration Steps.
The other path routes the signal to the marker circuits on the A50 board, where the signal is shaped into the other marker waveforms. The procedure for testing these two functions of the Tester are included here to demonstrate the operation of the dual voltage function of the Fluke A. The Calibrator Mainframe must complete a warm-up period and the SC must be enabled for at least 5 minutes prior to calibration to allow internal components to thermally stabilize.
For routine calibration, all steps except frequency and phase are necessary. Vertical deflection characteristics are verified by calibrating the voltage gain. The Event Status Register is a two-byte register in which the higher eight bits are always 0, and the lower eight bits represent various conditions of the A Calibrator.
You may need to adjust the signal averaging on the IB.
Fluke Calculator A/COIL User Guide |
If a specified instrument is not available, you can substitute an instrument that assures a 4: See Figure for test equipment connection instructions. Repeat the low frequency calibration procedure for the next amplitude unless the Calibrator Mainframe display indicates that the next steps calibrate pulse width.
Outguard Supplies The motherboard generates the outguard power supplies: Adjust the main time base position and vertical offset until the pulse signal is centered on the display. Typically this board shows aberrations of 0. Leveled Sine Wave Specifications Table Inguard Supplies The inguard supplies are located on the Voltage assembly A8.
If you make an entry error, press to clear the display, then reenter. Removing the Keyboard and Accessing the Output Block To remove the keyboard and access the output block, proceed as follows: Where To Go from Here The leveled sine wave signal is passed from the A50 board to the on-board attenuator assembly.
The blue softkey under Output toggles the impedance between 50 Q.
Fluke 5500a – Sc300 Multi-product Calibrator
Note this voltage or set to center of the display as it will be used as the reference for the following adjustments. Program the Calibrator Mainframe for an ffluke of 5. Center any aberrations so the peaks are equal above and below the reference level. Page – Amplitude Calibration Procedure for an O Trigger Signal Specifications Edge Function.